ا می طلبد
-ray radiation of low duty X-ray tube converted by primary radiation filters excites the fluorescence radiation of atoms of the sample under examination. Radiation flows (primary radiation scattered on the sample and secondary fluorescence radiation) comes in proportional gas meter and sulphur atom fluorescence radiation (SKa) is separated from radiation with other energies by means of selective filters.
The strength of sulphur atom fluorescence radiation recorded by proportional gas counter is proportional to the sulphur concentration in the material under examination.
|Method of sulphur concentration determination||Energy dispersive X-ray fluorescence (EDX) with selective filters|
|Statistic limit of detection, max., ppm||3|
|Range of determined sulphur concentrations, ppm||from 5 to 50000|
|Limits of basic relative error, %||±0,5|
|Power consumption, VA
(from mains 220 VAC, 50 Hz)
|Weight of the instrument, kg||12|
|Dimensions (L х W х H), mm||470х350х215|